Operational evaluation of the earlobe arterialized blood collector in critically ill patients
© Vaquer et al.; licensee BioMed Central. 2015
Received: 13 October 2014
Accepted: 26 February 2015
Published: 2 April 2015
The new Earlobe Arterialized Blood Collector (EABC®) is a minimally invasive prototype system able to perform capillary blood collection from the earlobe (EL) with minimal training and risk. This system could improve medical emergency management in extreme environments. Consequently, a prospective validation study was designed to evaluate operational performance of the EABC® in a cohort of critically ill patients.
Arterialized capillary blood was sampled from the EL of 55 invasively ventilated patients using the EABC® following a validated procedure. Operational characteristics such as the number of cuts and cartridges required, sampling failure/success ratio, bleeding complications, storage requirements and other auxiliary aspects were recorded. Result turnaround laboratory times (TAT) were compared with published references.
Blood collection was as easily performed on one earlobe as the other. Twenty-six minutes (mean 25.8; SD = 3.8) were required to obtain results, 15 min for patient preparation (mean 15.3; SD = 2.6) + 11 min for sampling and analysis (mean 11.4; SD = 2.1), which is similar to published hospital reference laboratory TAT. The average number of cartridges required was 1.3 (1–3; mode = 1) with the mean number of cut attempts being 1.2 (1–4; mode = 1). Problems/difficulties occurred in 59% of cases but were mainly attributed to patient’s demographic characteristics, with only 10% attributable to the collector (superficial cut, blood leak, collector misalignment and obstructed vision). Haemostasis was quickly achieved with minimum complications. Storage of the complete sampling kit required a 300 × 300 × 300 mm box. Two 9-V batteries were used during the 2-year study period.
The new EABC® system concept is safe, fast and easy to use. Observed problems/difficulties are easily amendable with certain design modifications. Definitive versions of the prototype have the potential for significant benefits for isolated and extreme environments in medicine.
KeywordsAcute respiratory failure Space medicine Capillary blood Intensive care medicine Mechanical ventilation
Critical medical situations are likely to occur during space missions in the future . In such situations, blood analysis capabilities, and especially arterial blood gasometrical evaluation, will be essential for proper medical management. Earlobe arterialized capillary blood gas analysis has been proposed as an adequate substitute for some arterial measures. This technique is based on achieving a capillary arterio-venous shunt in the earlobe so that capillary blood gas content approximates arterial levels . Nevertheless, discordant results are found in the literature on the accuracy of this technique, which has been extensively evaluated in healthy volunteers during exercise [3,4], in patients with different forms of respiratory failure [5-9], in patients under mechanical ventilation  and more recently in a meta-analysis .
Performance of a new prototype system for estimating arterial gas content from earlobe capillary arterialized blood, the Earlobe Arterialized Blood Collector (EABC®) + i-STAT® portable gas analyser, has been evaluated in healthy volunteers during head venous congestion, hypoxia and microgravity [12,13] and in patients with chronic renal failure . This new equipment minimizes room air contamination and enables standardization of sampling procedures. Results show good concordance of measures with arterial samples and much better safety profile, training requirements and patient tolerance than arterial sampling. It has not, however, been tested in critically ill patients. Therefore, a validation study was designed to evaluate its analytical performance and its operational requirements in a cohort of critically ill patients admitted to a general intensive care unit (ICU) for various conditions. Results from the analytical performance evaluation evidenced that the EABC® + i-STAT® is precise, capable of detecting extreme gasometric values in critically ill patients and could permit diagnosis and classification of patients presenting with acute respiratory distress syndrome (ARDS) . We present hereafter the operational evaluation of this equipment in the routine healthcare of critically ill patients.
Operational performance of the EABC® (Microgravity Centre/FENG-PUCRS, Brazil) + i-STAT® portable analyser (Abbot, United States of America) was evaluated in a cohort of adult patients admitted to a general ICU. Common demographic and clinical variables of study patients were recorded.
Two operators were trained on the technique and performed all evaluations. Training consisted of two separated workshops in which the producer presented the device and main procedures. Additional reference material and remote support were available as required after the practical sessions.
Informed consent by the patient or next of kin was required and local ethics committee accepted the study protocol (IRB: 2009593. Comité Ètic d’Investigació Clínica, Corporació Sanitària Universitària Parc Taulí, Sabadell, Spain).
Data were analysed using SPSS version 19 statistical software (International Business Machines. Armonk, New York. United States of America) and is presented as common statistical parameters (ratio, percentage, mean, mode and SD). Chi-Square test was used for evaluating qualitative variables. Two-tailed significance threshold was established at p < 0.05 for all statistical tests, and the 95% confidence interval is provided where available.
Study patient characteristics
Chronic cardiac failure
Diagnostic at ICU admission
The mean total time from decision to sample to results was 26 min (mean 25.8; SD = 3.8). Fifteen minutes (mean 15.3; SD = 2.6) were required for the preparatory phase and 11 min (mean 11.4; SD = 2.1) for blood collection and post-sampling procedures. The i-STAT® portable analyser required one automated calibration per sampling day which required 2–3 min.
EL arterialization with vasodilation cream and EL massage was performed during 2.5 min for the first 20 patients but was later increased to 5 min for the remaining 35 patients of the study. Earlobe blood congestion was achieved in both cases. Increasing arterialization time did not affect sampling success rate (OR = 0.86; p = 0.8; CI 95% = 0.28–2.67).
The average number cartridges required to obtain a measure was 1.3 (range = 1–3; mode = 1) and associated with a mean of 1.3 cut attempts (range = 1–4; mode = 1). Haemostasis was easily achieved with short-bleeding times (92.7% < 10 min) without complications (one case of suspected infection that quickly resolved spontaneously). Furthermore, no operator injuries or contamination by contact with patient’s blood were reported.
Auxiliary procedures for the i-STAT® portable analyser are described hereafter. Power consumption was low requiring one battery change (2 × 9-V battery) during the 2-year period of the study. Regular software updates were undertaken as required by the manufacturer. No additional auxiliary i-STAT® equipment was required for this study. Details on other i-STAT® technical specifications and performance characteristics can be found in the producer manual and are available online .
Storage space required for the i-STAT® portable analyser, automatic calibration system, computer-synchronization device, ten units of the EABC® and additional material for ten blood collections (sterile gloves, cleaning gauzes, etc.) fitted into a 300 × 300 × 300 mm box.
Cartridge shelf life if stored at room temperature (18°C–30°C) was 2 months; however, storage at 8°C increased lifetime until stipulated expiration date (minimum of 6 months).
The results from the present operational evaluation demonstrate that the EABC® + i-STAT® system concept is safe, fast and easy to use. The observed problems/difficulties attributable to the collector accounted for 10% of the cases and comprised of superficial cut, blood leak, collector misalignment and operator’s obstructed vision. These would be easily amendable with certain design modifications.
The results from the analytical performance evaluation have been published elsewhere  and showed good precision levels (coefficient of variation < 10% in all gasometrical variables) and enough accuracy to detect extreme gasometrical alterations of the EABC® + i-STAT® system (percentage of error < 20% in all gasometrical variables) when compared with direct arterial reference samples. This new system was also able to correctly classify ARDS patients according to severity (sensitivity = 100%, specificity = 92.3%, kappa coefficient = 0.85) . Therefore, the new technique was considered useful in the initial diagnosis and treatment of medical emergencies and could prove to be valuable if implemented in a medical protocol as a first evaluation tool of critically ill patients in remote environments. From an operational point of view, a limited sampling success rate was observed. The main causes of sampling failure were low-blood flow from the earlobe and blood coagulation, which had not been evidenced in previous validation studies using the EABC® + i-STAT® system [14,12,13] or when other capillary blood collection methods were used [3-10]. A separate evaluation was performed to ascertain what factors could influence earlobe capillary blood flow and affect sampling procedure, results of which have been published elsewhere . Using a multivariate analysis, which took into consideration several patient demographics and medical characteristics, the results evidenced that patient age was the only variable associated with sampling failure. It was hypothesized that microvascular ageing of the earlobe capillary bed could have influenced blood flow and blood delivery to the cartridge, leading to the observed low-sampling success ratio in older patients . Nevertheless, success rate was 90% in patients <50 years (n = 9) and 100% in 40-year-old patients (n = 4), which would support the use of the EABC® + i-STAT® system in a wide range of patients, especially those in isolated and extreme environments, such as astronauts in space  or explorers of remote locations on Earth.
The training required for learning the procedure of earlobe arterialized capillary blood sampling with the new EABC® + i-STAT® system was short, requiring only two workshops. Although a learning curve may exist, its interpretation was difficult and unreliable due to clustered patient distribution and was not used to determine training needs of the procedure. The two operators involved in the study reported that the sampling procedures were easy to perform in either of the patient’s earlobes. Furthermore, the system was found to be portable, manageable and efficient, requiring a reduced number of cuts and analysis cartridges in most of the cases. These findings confirm that the new system is easy to use and requires minimum training as reported in previous studies.
TAT is a key determinant of laboratory performance evaluation from an operational point of view . In the present study, a practical approach was chosen and TATs were calculated from the decision to sample until the finalization of post-sampling procedures. This estimation of TAT was considered to be the best interpretation and information of the real operational needs of the technique. Results showed that TAT of the whole procedure was 26 min, lying below current TATs of most tertiary hospitals . These results were considered adequate given the specific characteristics of the medical situations where the procedure could be used. However, since TAT calculation can be controversial sometimes , direct analysis time (or isolated laboratory TAT) can be evaluated. In this context, if preparatory procedures are excluded, results from blood gas analysis can be quickly obtained in 120 s using the i-STAT® portable analyser, which still represents an appropriate time to results . Furthermore, since earlobe arterialization was achieved in spite of increases in vasodilation time, further decrease in time expenditure for the whole procedure can be envisaged and could lead to a theoretical reduction of TAT to 23.5 min. Additionally, auxiliary procedures were found to require minimum time expenditure.
This study also confirmed the safety of the new system for both the patient and the operator. There was only one case of suspected infection of the EL, which resolved spontaneously. No episodes of accidental injury with the cutting blade or contamination of the operator with patient’s blood occurred. The safe use of the device was possible due to the containment of both blood and blade within the collector during operation. These findings are in agreement with previous studies in which no complications were reported.
The equipment utilized for providing autonomous gasometrical analysis capabilities was found to require a small storage volume. Furthermore, although the shelf life of a refrigerated cartridge can be extended to several months, room temperature storage during a maximum of 2 months is also possible. These findings make this new system very suitable for environments with limited storage capabilities such as the International Space Station and other space systems, as well as in extreme and isolated environments on Earth.
The present study has a number of limitations. First, the evaluation of the usability of the proposed new technique and collector comes substantially from the subjective opinion of operators, which could induce a bias in the evaluation. However, both operators were able to learn the procedure quickly and required only one cut to obtain a valid blood sample in most of the cases, which supports that sufficient skills on the use of the technique had been achieved. Second, as previously discussed, the presence of a learning curve may suggest that 35 sampling attempts are required to learn the procedure. However, unbalanced distribution of patients makes this assessment unreliable. There must be a minimum number of test rounds required; however, data are insufficient to determine such a number with precision. Third, this evaluation focuses on critically ill patients in the controlled environment of an ICU. While the use of a varied population of such patients may provide useful information that could be transferrable to other similar patients in extreme or remote environments, other uncontrolled factors such as temperature, humidity or atmospheric pressure may influence performance and operational characteristics of the technique. Therefore, additional field studies are required to confirm the potential benefits for patient healthcare in remote environments suggested in the present analysis.
In conclusion, this evaluation has demonstrated that operational characteristics of the new EABC® + i-STAT® system such as efficiency, sampling performance in specific patients, usability, minimum training requirements and reduced time to results, make it very suitable for use in remote and isolated environments such as space. Good accuracy and diagnostic capabilities in combination with abovementioned operational features suggest that definitive versions of the prototype can offer significant benefits for patient health care in space and other extreme environments. Future studies should focus on confirming the benefits of use of this new equipment and technique in such medical scenarios.
Earlobe arterialized blood collector
Intensive care unit
Acute respiratory distress syndrome
This study was funded by the European Space Agency via the Medical Projects and Technology Unit from the Crew Medical Support Office, European Astronaut Centre, Cologne (Germany). Funds were administered by the Fundació Parc Taulí, Sabadell (Spain) and delivered to the Critical Care Center, Corporació Sanitària Universitària Parc Taulí, Sabadell (Spain). We would like to thank all the aforementioned institutions for their economic and institutional support. Moreover, we want to recognize the significant contribution made to this project by the medical operations specialists of the Crew Medical Support Office (European Astronaut Centre, European Space Agency, Cologne, Germany) and give special thanks to the Centro de Microgravidade/FENG-PUCRS (Brazil) for its technical support throughout the development of this study.
- Billica RD, Simmons SC, Mathes KL, McKinley BA, Chuang CC, Wear ML, et al. Perception of the medical risk of spaceflight. Aviat Space Environ Med. 1996;67(5):467–73.PubMedGoogle Scholar
- Lilienthal JL, Riley RL. On the determination of arterial oxygen saturations from samples of ‘capillary’ blood. J Clin Invest. 1944;23(6):904–6.View ArticlePubMed CentralPubMedGoogle Scholar
- Fajac I, Texereau J, Rivoal V, Dessanges JF, Dinh-Xuan AT, Dall’Ava-Santucci J. Blood gas measurement during exercise: a comparative study between arterialized earlobe sampling and direct arterial puncture in adults. Eur Respir J. 1998;11(3):712–5.PubMedGoogle Scholar
- McEvoy JD, Jones NL. Arterialized capillary blood gases in exercise studies. Med Sci Sports. 1975;7(4):312–5.PubMedGoogle Scholar
- Dall’Ava-Santucci J, Dessanges JF, Dinh Xuan AT, Lockhart A. Is arterialized earlobe blood PO2 an acceptable substitute for arterial blood PO2? Eur Respir J. 1996;9(6):1329–30.View ArticlePubMedGoogle Scholar
- Eaton T, Rudkin S, Garrett JE. The clinical utility of arterialized earlobe capillary blood in the assessment of patients for long-term oxygen therapy. Respir Med. 2001;95(8):655–60.View ArticlePubMedGoogle Scholar
- MacIntyre J, Norman JN, Smith G. Use of capillary blood in measurement of arterial PO2. Br Med J. 1968;3(5619):640–3.View ArticlePubMed CentralPubMedGoogle Scholar
- Pitkin AD, Roberts CM, Wedzicha JA. Arterialised earlobe blood gas analysis: an underused technique. Thorax. 1994;49(4):364–6.View ArticlePubMed CentralPubMedGoogle Scholar
- Sauty A, Uldry C, Debétaz LF, Leuenberger P, Fitting JW. Differences in PO2 and PCO2 between arterial and arterialized earlobe samples. Eur Respir J. 1996;9(2):186–9.View ArticlePubMedGoogle Scholar
- Honarmand A, Safavi M. Prediction of arterial blood gas values from arterialized earlobe blood gas values in patients treated with mechanical ventilation. Indian J Crit Care Med. 2008;12(3):96–101.View ArticlePubMed CentralPubMedGoogle Scholar
- Zavorsky GS, Cao J, Mayo NE, Gabbay R, Murias JM. Arterial versus capillary blood gases: a meta-analysis. Respir Physiol Neurobiol. 2007;155(3):268–79.View ArticlePubMedGoogle Scholar
- Russomano T, Evetts SN, Castro J, Dos Santos MA, Gavillon J, Azevedo DFG, et al. A device for sampling arterialized earlobe blood in austere environments. Aviat Space Environ Med. 2006;77(4):453–5.PubMedGoogle Scholar
- Russomano T, Ernsting J. The arterialized ear lobe blood samples for blood gas tensions - a technique for medical emergencies in microgravity. Beijing, China: IAF Proceedings, 47th IAC Congress; 1996.Google Scholar
- Falcão F, Russomano T. Clinical validation of the earlobe arterialized blood collector. Aviat Space Environ Med. 2010;81(11):1053–4.View ArticlePubMedGoogle Scholar
- Vaquer S, Masip J, Gili G, Gomà G, Oliva JC, Frechette A, et al. Earlobe arterialized capillary blood gas analysis in the intensive care unit: a pilot study. Ann Intensive Care. 2014;4:11.View ArticlePubMed CentralPubMedGoogle Scholar
- Abbott Point of Care. Performance Characteristics of the i-STAT® 1 Portable Analyzer, in i-STAT® 1 System Manual. 2012.Google Scholar
- Barratt MR, Pool SL, Eds., Principles of Clinical Medicine for Space Flight (1st ed.). New York: Springer-Verlag. Various Chapters. 2008.
- Hawkins RC. Laboratory turnaround time. Clin Biochem Rev. 2007;28(4):179–94.PubMed CentralPubMedGoogle Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.